Q: How do I order an amalgam separator or replacement container?
Click HERE for Separators
Click HERE for Replacement Containers
Part numbers are as follows:
NXT Hg5 Amalgam Separator, NXT-HG5-001 (1-10 chairs)
NXT Hg5 mini Amalgam Separator, NXT-HG5-MINI (1-4 chairs) – suitable for tight spaces only
Hg5 High Volume Amalgam Separator, HG5HV-001 (11-20 chairs)
NXT Hg5 Replacement Collection Container with Recycle Kit, NXT-HG5-002CR
Hg5 Replacement Collection Container with Recycle Kit (Legacy), HG5-002CR
NXT Hg5 Recycle Kit Only (does not include a new collection container), NXT-Hg5-002R
Hg5 Recycle Kit Only (does not include a new collection container), Hg5-002R
Q: What is the difference between the NXT Hg5, the NXT Hg5 mini and the Hg5 High Volume?
A: The NXT Hg5 system is designed to handle between 1-10 chairs. The inlet and outlet measure 1 ½”
and it can be installed on the floor or wall mounted.
The NXT Hg5 mini is designed to handle 1-4 chairs, has ¾” inlet and outlet, and was originally intended
to be installed under a counter or lab sink. But the mini can also be wall mounted or stand on the floor.
The mini is 7.5” shorter than the standard NXT Hg5.
The Hg5 High Volume is a system designed to handle 11-20 chairs. The High Volume can only be wall
mounted and it has 3” inlet and outlet. This system has 2 containers that will eventually need replacing.
Q: What collection container should I order for my system?
A: First, determine which Amalgam Separator you are currently using. The latest generation in the
Hg5 Series of Amalgam Separators is the called the NXT Hg5 Amalgam Separator (released in June
2017). If purchase was made prior to June 2017, then you may be using the Hg5 Amalgam
Separator. The following shows which collection container you should order:
NXT Hg5 Amalgam Separator – use the NXT Hg5 Collection Container (NXT-Hg5-002CR)
NXT Hg5 mini Amalgam Separator – use the NXT Hg5 Collection Container (NXT-Hg5-002CR)
Hg5 Amalgam Separator (Legacy) – uses the Hg5 Collection Container (Hg5-002CR)
Hg5 mini Amalgam Separator (Legacy) – uses the Hg5 Collection Container (Hg5-002CR)
Hg5 High Volume Amalgam Separator – uses the Hg5 Collection Container (Hg5-002CR)
Q: What if I neglected to change my container and now there is water/sediment in the top
A: First and foremost, you should put a new collection container on the system as soon as possible.
After that, depending on the level and volume of the material in the top chamber, you have a few
options that can help clear the top chamber.
IF Low Level of Material:
1. Leave suction ON.
2. While holding the collection container in place, remove the 2 pins that attach the container to the top
3. Slowly pull down on the collection container to break the seal between the top and bottom container.
As soon as air gets inside the top chamber you should see the contents start to agitate. Allow the
material to agitate for 30 seconds or so, then put the container back into place and secure with the
pins. The material in the top should start to flow to the bottom container almost immediately.
(Sometimes this procedure is required to be done a few times before all the top contents have moved
to the bottom.)
IF Medium/High Level of Material:
1. Try the above process first. If that doesn’t work, you should call a service technician to try and help
you clear the top. Keep in mind, the waste collected in the top chamber should be recycled, just like
the collection container. So if your service technician manages to get all the waste out of the top and
into a bucket, that bucket should be sent off to a mercury recycler.
2. If emptying the top chamber into a bucket is not an option, you can purchase a new air-water
separator (top) from your dental supplier. The part number for that item is NXT-HG5-AWSEP. Please
keep in mind that the full top chamber you are replacing still needs to be sent to a certified mercury
Q: What if I have a portable vacuum? Can an amalgam separator be installed on the cart?
A: Unfortunately, portable vacuums are not designed the handle the size and weight associated with
most amalgam separators on the market. However, a separator could be placed in/next to a sink, and
the contents of the portable vacuum could be poured into the inlet of the separator, with a hose or tube
leading the outlet of the separator to the sink. This would require gravity to do the work, instead of
suction, but over time it would still allow the material to settle and the separator to do its job. With the
proper shut off valve, the amalgam separator could be taken off the sink and placed in storage until the
next time it is needed.
Q: I don’t have central suction. Can one of the Hg5 series of amalgam separators work for me?
A: If you currently don’t have central suction, and work off a Venturi system, unfortunately our line of
Hg5 separators would not accommodate your system.
The NXT Hg5® is designed to provide years of trouble-free service, with minimal attention. In the
unlikely event of system related problems, please consult the troubleshooting and maintenance chart in
your Hg5 manual, or call SolmeteX® for assistance.
Low Vacuum at the Hand Piece
If the vacuum gauge reads normally but there is little or no vacuum to the hand piece, there is probably
a clog or a leak between the hand piece and the NXT Hg5® Amalgam Separator.
If the vacuum gauge reads lower than normal these are the possible causes:
Q: I just replaced the collection container. Why is my vacuum LOW?
Check installation of the new collection container. If the vacuum is low there is a possibility that the o-
rings on the collection container did not seal into the receivers. Remove the collection container, check
the o-rings and re-install per the collection container installation procedure.
1. Check solid collection cup(s) – Clean or replace element/screen if necessary.
2. Check operation of the vacuum pump.
3. Listen for vacuum leaks.
4. Check all connections for breaks or cracks.
5. Check all flexible hose for kinks, breaks, or loose clamps.
Practice Waste Solutions – Waste Compliance
Q: Some states say that storage of waste amalgam cannot exceed 1 year. Does that include my
amalgam separator container?
A: As long as the container is actively being used as part of the system, it is not considered “storage”.
If you were to take the container off, and sit it on the floor, that would be considered “storage” and you
would have 1 year to make sure it is properly recycled.
Q: Can sharps go in my biohazard bucket?
A: Loose sharps are not permitted in the biohazard bucket. However, full sharps containers may be
placed in the biohazard bucket.
Q: Why does California require a different filter for the fixer & developer waste?
A: Although developer does not meet the Federal requirement for silver discharge limits, there are
several counties in California that require fixer AND developer to be filtered prior to discharge. Water
treatment facilities in parts of CA have trouble with their pH levels, therefore they require dental offices
to treat their fixer and developer to help control pH levels being put into their facilities.
Q: What is the difference between an automatic and manual filter for fixer recovery?
A: Dental offices either have an automatic film processor, in which the processor is plumbed directly to
a drain, or they have a manual processor that requires emptying of a tray or bottle manually after use.
Q: What kind of amalgam waste can into the amalgam buckets?
A: The amalgam buckets are designed to handle contact and non-contact amalgam waste generated in
dental practices. That includes vacuum trap filters, chair side traps, extracted teeth with amalgam, and
spent amalgam capsules. Liquid mercury cannot be put in an amalgam bucket. If you have liquid
mercury, and you need help in disposing of it, please call Solmetex directly at 800-216-5505.
Q: What kind of waste can be put into a scrap lead bucket?
A: Our lead bucket is designed to handle lead foil from x-ray films and auto clave indicators. We offer a
separate product for recycling lead aprons. The Lead Apron Bucket (PWS-LB-A) is a bucket designed
to hold 2 lead aprons. Lead aprons must be recycled separately from lead foils
Q: How long can I store biohazard waste in my office?
A: Each state has its own storage limits for bio-hazard waste. Please click here to find your state and
Copyright © 2017 SolmeteX